<!--
 * @Author: cgl
 * @Date: 2023-04-18 15:12:26
 * @LastEditors: cgl
 * @LastEditTime: 2023-09-04 15:23:28
 * @FilePath: \public-health-emergencies-fed:\新放射\新radiation-hygiene-fe\src\views\person\itemControl\item12.vue
 * @Description:
 *
 * Copyright (c) 2023 by cgl/微盛信息产业有限公司, All Rights Reserved.
-->
<template>
	<div>
		<table border="1" cellspacing="0" cellpadding="0" align="center" style="width: 100%;">
			<tr class="title">
        <td rowspan="3">序号</td>
        <td rowspan="3">检测项目</td>
        <td colspan="4">指标要求</td>
			</tr>
			<tr class="title">
        <td>验收检测</td>
        <td>状态检测</td>
        <td colspan="2">稳定性检测</td>
			</tr>
      <tr class="title">
        <td>判定标准</td>
        <td>判定标准</td>
        <td>判定标准</td>
        <td>检测周期</td>
      </tr>
      <tr>
      	<td>1</td>
      	<td>源活度</td>
      	<td>±5%</td>
      	<td>±5%</td>
      	<td>±5%</td>
      	<td>换源或维修后(<i>$^{198}Ir$</i>),<br/>三个月(<i>$^{60}Co$</i>)</td>
      </tr>
      <tr>
      	<td>2</td>
      	<td>源传输到位精确度</td>
      	<td>±1mm</td>
      	<td>±1mm</td>
      	<td>±1mm</td>
      	<td>换源或维修后(<i>$^{198}Ir$</i>),<br/>三个月(<i>$^{60}Co$</i>)</td>
      </tr>
      <tr>
      	<td>3</td>
      	<td>放射源累计定位误差</td>
      	<td>±2mm</td>
      	<td>±2mm</td>
      	<td>—</td>
      	<td>—</td>
      </tr>
      <tr>
      	<td>4</td>
      	<td>贮源器表面(5cm、<br/>100cm)泄漏辐射所<br/>致周围剂量当量率</td>
      	<td>5cm(50μSv/h);<br/>100cm(5μSv/h)</td>
      	<td>5cm(50μSv/h);<br/>100cm(5μSv/h)</td>
      	<td>5cm(50μSv/h);<br/>100cm(5μSv/h)</td>
      	<td>换源或维修后(<i>$^{198}Ir$</i>),<br/>三个月(<i>$^{60}Co$</i>)</td>
      </tr>
      <tr>
      	<td>5</td>
      	<td>源驻留时间误差</td>
      	<td>±0.5s</td>
      	<td>±0.5s</td>
      	<td>—</td>
      	<td>—</td>
      </tr>
      <tr>
      	<td>6</td>
      	<td>多源系统重复性</td>
      	<td>0.02%(源单独选择多<br/>源系统);0.03%(源随<br/>机选择多源系统)</td>
      	<td>0.02%(源单独选择多<br/>源系统);0.03%(源随<br/>机选择多源系统)</td>
      	<td>—</td>
      	<td>—</td>
      </tr>
      <tr>
        <td colspan="6" style="text-align: left;">注:剂量重复性检测无周期要求,但在投入使用前或大修后应做该项检测</td>
      </tr>
		</table>
	</div>
</template>

<script>
export default {
  name: "item41",
  data() {
    return {
    };
  },
  created() {
  },
  methods: {
  }
};
</script>

<style lang="scss" scoped>
@import './content.scss';
</style>
